Philips discovered that some welding joints on the vertical carriage brackets of the tube stand may crack. As a result, the tube arm assembly may exhibit uncharacteristic sideways movement, with the tube becoming off center. If all of the welding joints cracked, the tube arm assembly could fall down.
Lot or serial number
Model or catalog number
- PHILIPS HEALTHCARE (SUZHOU) CO., LTD
NO. 258, ZHONG YUAN ROAD